1. Airflow resistance test
2. Adsorption efficiency/penetration rate
3. Determine desorption
Meet and comply with the following regulations
·Online Dust Particle Counter Calibration Specification- Draft for Review·ISO21501-4
: 2007 "Single Particle Optical Measurement of Particle Size Analysis Part 4 Clean Room Light Scattering Dust Particle Counter"
·GB/T6167-2007 "Dust Particle Counter Performance Test"
· JJF1190-2008 "Dust Particle Counter Calibration Specification"
·JIS B9921-2010 "Clean Room Light Scattering Dust Particle Counter"
This device is composed of aerosol generator, mixing system, flow reference and concentration calibration system. It uses the aerosol generation system to provide a quantitative supply of test particles, and uses uniform dilution, diffusion, stabilization, etc. to transport aerosols with a set concentration, and adopts aerosol photometer mass concentration standard device (reference photometer) or filter film to weigh Law as a traceability method.
|Calibration instrument||2.83L/min, 28.3L/min, particle counter/sensor|
|Occurring particle size range||(0.3～10)μm|
|Number of channels||2|
|Particle concentration range||(3000～30000)pcs/28.3L|
|Reference Size||330mm (length) × 400mm (width) × 370mm (height), mixing distributor 230mm (diameter) × 920mm (height)|
·Dust particle sensor repeatability;
·Dust particle sensor particle concentration indication error;
·Dust particle sensor particle size distribution error;
·Dust particle sensor flow rate indication error;
No.58 Zhengliang 4th Road, Daoyi Development Zone, Shenbei New District, Shenyang City
Official website of mobile phone
WeChat official account
Copyright © 2021 xinke.net.cn All rights reserved. 辽ICP备05024146号-1 Powered by：300.cn Shenyang